Apparatus for applying a Micro-needle Therapy by vibration.
A device for puncturing skin-penetrating drugs or cosmetics supports with a micro-needle guided between extended and retracted
-Type of protection against electric shock: Class II Equipment
-Degree of protection against electric shock: Floating Applied Part
-Degree of protection against the ingress of
-Mode of operation: Continuous operation
REGION THIN SKIN THICK SKIN
Forehead 0.25 0.5mm 0.5 0. 75mm
Between Eyebrow 0.25 0.5mm 0.5-1.0mm
Nose 0. 25mm 0.5mm
Eye Area 0.25mm 0.25 0.5mm
CheeK Bone 0.5mm 0.5 1.0mm
Check and Chin 0.5 1.0mm 1.0 2.0mm
Upper UP 0.25mm 0.25 0. 75mm
Facidl Scarring 1.0 1.5 2.0mm
Stretch Mark 1.5 2. 0mm 1. 5 2.0mm
Surgical k Burn Scars 2.0 2. 5mm
1.General -Keep it with its power off. Wipe its surface out with a dry cloth.
-Keep it away from direct sunlight or high temperature and humidity.
-Do not leave the measurement, repair or alter of the device to non-professional staff.
-Make sure to inspect the device and adapter before use.
-When you use the device which has not been used for a long time, make sure to check if il works properly with safety.
(1) plug in DC adaptor to the socket. (2)Plug in DC adaptor to AC outle
(1)Take out needle cartridge in the sterilized packing.
(2)Fitting round slots 1 into 2 of machine, push needle cartridge horizontally up to the end.
(3) Lock the cartridge by turning it to the counter clockwise.
(4)Adjustmentof needle length should be done after turn on the machine.
(5)lf you want to remove the needle cartridge, turn il to Ihe clockwise and push out horizontally, fitting round slots(s!ot1.slol2) together.
Rotate the speed ring to adjust the speed. 1 The speed will be faster if you turn the ring clockwise.
1 The speed will be slower if you turn the ring counterclockwise.
Adjustment of needle length
You can adjust the length of needle by turning the adjustment ring clockwise or counter clockwise.
Turning it clockwise, length of it will be longer.
Turning it counter clockwise, length of it will be shorter.
There are no reviews yet.